Downloads by Zenith
GPIB Communicator 1.91 (Jan 30, 2017)
ScopeWFM 1.2 (Oct 9, 2006)
Set PS 1.3
(Oct 17, 2015)
Set FG 1.1 (Feb 10, 2017)
Set SI 5010 1.0 (Jan 22, 2016)
Show DM 1.1 (Jan 26, 2015)
TDS Performance Verification (Sep 10, 2017)
TDS Error Log Viewer 1.23 (Aug 27, 2017)
Measurement Logging 1.41 (March 10, 2013)
Tek Parts Manager 1.23 (June 24, 2012)
HP Parts Manager 1.0 (Jan 9, 2010)
2400DSO Quick Check 2.0 (March 29, 2013)
Compare GPIB Command Files 1.0 (June 20, 2015)
Check Terminations and Attenuators 1.3 (March 15, 2017)
OT, Folder Copier 1.61 (May 30, 2017)
OT, File Finder 2.57 (Sep 20, 2017)
Obsolete Tek Softwares:
XYZ's of Oscilloscopes, Windows helpfile version of the Tek document with the same name.
Docuwave 1.2, Windowsstyle DOS program that imports waveforms from numerous Tek scopes like 11k, 2200, 2400, TDS and more.
Grabber 1.0, DOS program that imports waveforms from 2201.
WaveStar Lite 1.0, Windows program that imports waveforms from TDS 200-series.
TEKCATS 11300, Adjustment software v2.19 (DOS). Unzip and run install.bat.
TEKCATS DSA600, Adjustment software v1.2 (DOS). Unzip to the root directory of any drive. Rename the directory to TEKCATS.
SD ID utility, Small DOS software to enter the serial number into an SD2x series sampling heads.
Troubleshooting your Oscilloscope
As the title says this is a book on troubleshooting older oscilloscopes. I found it some time ago among my other manuals and scanned it.
It was probably a handout from one of the many Tek Service training courses I have attended.
This document is often found for sale on eBay by sellers who downloaded it for free, either from here or from some other free site.
I can tell it's mine since it's Letter size and it was scanned in A4 size leaving a white band on the bottom as seen on the pictures.
This is a rescan in the correct Letter format.
Another handout from a training course. Kind of basic but maybe useful anyway.
A Training Information Brochure which was sent to Field Offices to familiarize 7k technicians with new techniques used in the new 7104 scope.
Troubleshooting tips on 2200-series power supply
Troubleshooting tips on TDS300-series power supply
Documents I wrote based on my own experience working on these scopes for many years.
1969 Catalog Supplement introducing the 7000 series
Parts Replacement kits, Mod kits, Data Sheets and other Documents
Scans of various documents
Home made TM500 Plug-Ins
Modification of an SI 5010
Installing an LCD display in a TDS544A
Building a dual fast pulse generator
Modification of a DC5010 EXT Clock Input
Replacing the V/Div Scale Factor lamps with LED's in a 465
Building an RF Step Attenuator
Recapping a 7603
Convert a 067-0587-02 to 067-0587-10 for use in 11k
Fix for a common problem on SG5030 levelling heads
A comparison between a low cost Chineese 500 MHz probe and a Tek P6139A
Surplus Instruments, Boards & Parts
Picture of my lab